 Methodology
 Open Access
 Published:
Advanced modes of mechanical ventilation and optimal targeting schemes
Intensive Care Medicine Experimental volume 6, Article number: 30 (2018)
Abstract
Recent research results provide new incentives to recognize and prevent ventilatorinduced lung injury (VILI) and create targeting schemes for new modes of mechanical ventilation. For example, minimization of breathing power, inspiratory power, and inspiratory pressure are the underlying goals of optimum targeting schemes used in the modes called adaptive support ventilation (ASV), adaptive ventilation mode 2 (AVM2), and MIDfrequency ventilation (MFV). We describe the mathematical models underlying these targeting schemes and present theoretical analyses for minimizing tidal volume, tidal pressure (also known as driving pressure), or tidal power as functions of ventilatory frequency. To go beyond theoretical equations, these targeting schemes were compared in terms of expected tidal volumes using different patient models. Results indicate that at the same ventilation efficiency (same PaCO_{2} level), we expect tidal volume dosage in the range of 7.4 mL/kg (for ASV), 6.2 mL/kg (for AVM2), and 6.7 mL/kg (for MFV) for adult ARDS simulation. For a neonatal RDS model, we expect 5.5 mL/kg (for ASV), 4.6 mL/kg (for AVM2), and 4.5 (for MFV).
Background
Adaptive ventilation modes are designed to automate some of the basic actions of clinicians as they attempt to identify the best settings, although the definition of “best” continues to be a matter of debate. The algorithms usually adapt to the changing characteristics of the patient, such as mechanics (resistance, compliance, and inspiratory effort) or ventilatory pattern (frequency and tidal volume), and choose an appropriate response. One strategy to incorporate clinical knowledge into machine design is to use what is called an optimum targeting scheme [1], a term adapted from engineering control theory. An optimum targeting scheme is based on a mathematical model that attempts to minimize or maximize some desired outcome. In optimization theory, that model is also called a cost function. This function tells the machine how much a ventilation pattern “costs” in terms of predefined criteria. These criteria are based on actual patient characteristics (e.g., the cost function could simply describe tidal volume dosage). Hence, the goal of an optimum targeting scheme is to find the ventilation pattern with the lowest cost. If this optimum pattern is found, it can be used to set values (targets) for the underlying controllers. For a detailed description of how optimal targeting schemes work, see the Additional file 1.
In the first section of this paper, familiar ventilation parameters (tidal volume, tidal pressure, and tidal power) are used to derive cost functions. Also, the cost functions, which underlie the ventilation modes called adaptive support ventilation (ASV), adaptive ventilation mode 2 (AVM2), and mid frequency ventilation (MFV), are derived. Next, we perform mathematical analyses to compare the characteristics of these optimum target schemes. These analyses tend to be complex and hard to interpret intuitively. For this reason, a summary of clinical evidence is presented along with mathematical simulations we performed to compare and visualize the results of the cost function minimization.
Figure 1 shows the variables used to derive the cost functions and how cost functions are used to optimize the ventilatory pattern.
Methods
What should we optimize?
Tidal volume
The pivotal study by the Acute Respiratory Distress Syndrome Network in 2000 established the notion that in patients with acute lung injury and acute respiratory distress syndrome, mechanical ventilation with a lower tidal volume dosage (6.2 vs 11.8 mL/kg ideal body weight) decreases mortality and increases the number of ventilatorfree days [2]. There are also data to support the use of low V_{T} in patients without preexisting lung injury [3,4,5,6]. A recent study even suggests that lung protective ventilation might be considered a prophylactic therapy, rather than just a supportive therapy [7].
If we assume a value for the required alveolar minute volume (MV_{A}) and simply desire to control the tidal volume (V_{T}) dosage for a passive patient, we can derive the cost function as follows:
where V_{D} represents the dead space volume and f the ventilatory frequency, and hence, MV_{A}/f represents the alveolar volume. Thus, the “cost” in terms of tidal volume dosage (and presumably the risk of VILI) goes down as frequency goes up for a given required minute alveolar ventilation. However, we see that there is no definite minimum value because tidal volume converges to the dead space volume as frequency increases to infinity. In practice, the limit would be dependent on the volume delivery performance characteristics of the ventilator, because no ventilator is a perfect flow controller. Also, in the USA, conventional ventilator frequency is limited to a maximum of 150 breaths/minute.
Tidal pressure
Simply controlling the tidal volume dosage, independent of any consideration of lung mechanics, may have limited utility. Recent work has suggested that V_{T} normalized to lung mechanics (e.g., V_{T}/C) is a better predictor of mortality than tidal volume dosage [8,9,10]. We prefer to call V_{T}/C (or equivalently, P_{plat} – totalPEEP) tidal pressure, P_{T}, instead of driving pressure because P_{T} differs from V_{T} by only a scaling factor and driving pressure is sometimes used in reference to any pressure driving flow, not just static endinspiratory pressure at the airway opening. In a cohort of braininjured patients, P_{T} was associated with the development of ARDS [11]. In a series of ARDS patients receiving ECMO for refractory hypoxemia, P_{T} during ECMO was the only ventilator setting that showed an independent association with inhospital mortality [12]. In patients having surgery, intraoperative high P_{T} and changes in the level of PEEP that resulted in an increase of P_{T} were associated with more postoperative pulmonary complications [13]. On the other hand, if V_{T} is strictly maintained at 6 mL/kg predicted body weight and P_{plat} below 28–30 cm H_{2}O, then P_{T} shares the same information as P_{plat} about the association with day 90mortality [14].
However, if we define an optimal targeting scheme as minimization of P_{T}, we get the same result as minimizing to tidal volume because the tidal pressure is linked to driving pressure by compliance, C, which can be considered simply a scaling factor. If compliance only affects the scaling of the cost function, then it has no influence to the location of the minimum.
Tidal power
Gattinoni et al. have suggested an association between power transfer (from ventilator to lungs) and VILI [15]. However, as Marini and Jaber have observed [16] “…it is difficult to link power dissipated in proximal airway resistance directly to noxious events at the alveolar level.” Furthermore, they discount the effect of PEEP on the power equation because “…the ventilator’s work against PEEP is temporarily stored as potential energy within the elastic tissues of the respiratory system; it later converts to kinetic energy as the gas escapes to atmosphere across the exhalation valve.” Hence, the power used to deliver the tidal volume against PEEP is not stored in the body and would not be expected to contribute to lung injury. Thus, they have suggested that a potentially better indicator of injury risk for clinical purposes might be “driving power” defined as:
where C (compliance) is a scaling factor used to account for “…the reduced capacity of the ‘baby lung’.” However, we can define tidal power as
which is equal to total power without the resistive portion and the energy which escapes to atmosphere during expiration. Marini and Jaber suggested driving power as a metric that could be associated with the risk of VILI and recommended that power be normalized “…at least for aerated lung capacity.” If tidal power is used as the cost function, we replace V_{T} in Eq. (4) with Eq. 1 and then solve the following optimization problem:
The solution is obtained analytically by differentiating tidal power with respect to f and setting the result to zero. Solving Eq. 5 for optimal frequency leads to the remarkable result of
where f_{TP} = frequency of minimum tidal power and MV = minute volume measured at the proximal airway. If we express MV as the product of tidal volume and frequency, the optimal tidal volume (i.e., optimal in terms of minimal tidal power) can be expressed simply as function of dead space:
Furthermore, if we assume V_{D} = 2.2 mL/kg (IBW) as an estimation for normal dead space volume, the tidal volume would be given by:
for minimal tidal power to ventilate normal lungs. As mentioned, driving power is connected by a scaling factor to tidal power. Therefore, the condition of minimal driving power is fulfilled at the same optimal frequency and therefore yields the same optimal tidal volume.
Cressoni et al. defined transpulmonary mechanical work as the area between the inspiratory limb of the transpulmonary pressure vs volume curve during inspiration with constant flow [17]. They showed that if transpulmonary mechanical power (work per breath times respiratory frequency) exceeded the limit of 12 J/min, five out of five piglets developed wholelung edema and four out of four did not when they were ventilated below that threshold.
Breathing power (Adaptive Support Ventilation)
In 1950, Otis et al. investigated unassisted breathing frequency with respect to lung mechanics and alveolar minute ventilation [18]. They made the assumption that the brain seeks an optimum frequency by minimizing breathing effort. To derive the cost function of breathing effort, they assumed a one compartment lung model with linear compliance and nonlinear resistance:
where R is the linear (viscous) and R′ the nonlinear (turbulent) portion of airway resistance and P_{mus} the pressure generated by inspiratory muscles. The flow \( \dot{V} \) was assumed to follow a sine curve with an I:E ratio of 1:1:
where \( \widehat{\dot{V}} \) represents the peak flow. On the basis of that model, breathing effort was defined as work rate or power. With the assumptions of Eqs. 9 and 10, the mean rate of muscular work was derived as [18]:
To find the optimal frequency at minimal breathing power, the following optimization problem must be solved.
Otis solved Eq. 12 by differentiating Eq. 11 with respect to f and setting the result equal to zero. Instead of solving for f, he solved the equation for MV_{A} to get a solution for the conditions of minimal breathing power. Later, Mead [19] simplified Eq. 9 by neglecting the term with turbulent flow resistance (\( {R}^{\prime}\cdot \dot{V}{(t)}^2 \)). Then, he solved the optimization problem of Eq. (12), resulting in the wellknown equation for determining the optimal frequency at minimal breathing power (f_{BP}):
As an alternative to Eq. (13), Mead also showed that an optimal frequency exists at which the average force per breath required from the respiratory muscles is minimal (f_{BF}):
Otis et al. and Mead derived their equations to better understand the energetics of breathing and the associated effects on “the imaginary path from health to disease.” They were not concerned with inventing new modes of mechanical ventilation.
In 1991, Fleur T. Tehrani patented a targeting scheme based on Eq. 13. The system was designed to “…reduce the load on the respiratory muscles, mimic natural breathing, stimulate spontaneous breathing, and reduce weaning time” [20]. Interestingly, the initial implementation of this targeting scheme was not to minimize power delivery from ventilator to patient [21], but rather to select initial settings and “…choose a breathing pattern that encourages the patients to breathe on their own as early as possible” [22]. Note that the development of this targeting scheme was almost a decade before intensive research on the role of tidal volume dosage on mortality. At that time, the concern was to avoid an excessively large tidal volume, not to minimize it. Nevertheless, over the years, ASV has proven to be effective and results in relatively protective tidal volume delivery in the range of 8.1 ± 1.4 mL/kg ideal body weight [23].
Inspiratory power (Adaptive Ventilation Mode 2)
Ventilation modes using adaptive targeting based on Eq. 13 do not necessarily deliver lung protective ventilation [24, 25]. To reduce tidal volume (and subsequently tidal pressure) [10], we can derive the concept of mean inspiratory power [26]. Inspiratory power is defined as the sum of the resistive and tidal power which is transmitted from the ventilator to the patient assuming intrinsic PEEP equal zero:
There are differences among inspiratory power, total power [15], elastic power, breathing power, and tidal power. Elastic power includes tidal power and PEEP power, inspiratory power includes tidal power and resistive power, and total power includes elastic power and resistive power. Figure 2 and Table 1 explain these concepts (which were created by Otis, Gattinoni, Marini, and us). Note that power is defined as the work per unit time, which is calculated as the product of work and ventilatory frequency. Inspiratory work per breath is defined as the integral of inspiratory pressure with respect to inspiratory volume, or graphically, the area between the pressure curve and the volume axis as shown in Fig. 2.
There is an important difference between muscle power \( {\dot{W}}_{\mathrm{mus}} \) and inspiratory power. Otis derived the mean power which is needed to breathe without the support of a ventilator with a sinusoidal muscle pressure waveform. On the contrary, the concept of inspiratory power relies on the principle of how much power is delivered to the patient by a ventilator using a square pressure waveform (assuming total PEEP equals zero).
Inspiratory power is not intended to be another predictor for VILI. Instead, it serves as the basis for defining an alternative cost function which may be used to describe an optimal ventilation pattern. Inspiratory power includes not only tidal power (which might be a better indicator for VILI) but also includes resistive power. This leads to a more “natural” ventilation similar to Otis’ breathing power. However, as we will see, minimizing inspiratory power converges for specific patient characteristics to the same result as minimizing tidal power which might be relevant for VILI prevention.
The derivation of inspiratory power for pressure controlled ventilation of a onecompartment linear lung model is provided in the Additional file 1. Inspiratory power can be calculated as:
where coth() is the cotangens hyperbolicus function and T_{I} is the set inspiratory time on the ventilator. To find the optimal frequency for minimal inspiratory power f_{IP}, the following optimization problem must be solved:
Assuming that I:E = 1:1, the following numerical solution can be derived:
Note that Eq. 18 is a socalled “fixed point iteration.” That means we cannot directly calculate the optimal frequency f_{IP}. The optimum frequency is found using an iterative numerical process, starting with a seed value. It can be shown that Eq. 18 converges to the solution for minimal tidal power of Eq. 6 for small respiratory system time constants (see Additional file 1). Therefore, the optimal frequency for minimal inspiratory power is always equal to or less than the frequency for minimal tidal power.
Inspiratory pressure (Mid Frequency Ventilation)
Marini et al. derived an equation which allows prediction of tidal volume in terms of ventilator settings and lung mechanics [27]. In 2013, Chatburn and MirelesCabodevila extended this equation to predict alveolar minute volume as a function of frequency and invented a new optimal targeting scheme called midfrequency ventilation (MFV) [28]. MFV is designed to maximize alveolar minute ventilation for a given inspiratory pressure target [28] or minimize inspiratory pressure target for a target minute alveolar ventilation [29] (inspiratory pressure target, P_{insp}, is the preset pressure change above PEEP set on the ventilator, i.e., the amplitude of the square pressure waveform).
According to Marini, tidal volume can be expressed as:
where R_{I} is the inspiratory resistance, R_{E} the expiratory resistance, and D the fraction of Ti to the period T. The ventilation frequency f is denoted in hertz. With a constant dead space volume V_{D}, the alveolar minute volume is given by:
To get the optimal ventilation frequency, the following equation must be solved to maximize alveolar minute volume:
This can be solved experimentally by trying out different frequencies [28]. Equation 21 can be also used in a different way to minimize inspiratory pressure under the condition of constant minute volume. Hence, it can be used in the same way as the equations of ASV or AVM2. Consequently, we are able to find the minimal required set inspiratory target pressure for a desired alveolar minute volume. The frequency of minimal inspiration pressure (P_{insp}) can be found by solving the following optimization problem:
where P_{insp} can be expressed by rearranging Eq. (21) as:
Equation 23 can be also solved experimentally by substituting different values for ventilation frequency in Eq. 24. Logically, the same frequency which maximizes alveolar minute volume will also minimize inspiratory pressure.
Evidence for optimum targeting schemes
Note that the Additional file 1 contains extensive tables summarizing the published evidence in simulations, animal studies, and human studies.
Much has been written about ASV (note that IntelliVent ASV is an advanced variety of ASV, with the same taxonomic mode classification but with the addition of automatic control of minute ventilation target, PEEP, and FiO_{2}). In preparing this manuscript, a Google search using the term “Adaptive Support Ventilation” revealed 72 references between 2000 and 2017. ASV evolved as a form of the mode called Mandatory Minute Volume described by Hewlett et al. in 1977 [30].
MirelesCabodevila and Chatburn [28] introduced MFV in 2008. They used an interactive mathematical model of ventilator output during pressure control ventilation (implemented as MFV) to predict the frequency at which alveolar ventilation is maximized with the lowest tidal volume for a given inspiratory pressure target. The results of the mathematical simulation were verified with a mechanical breathing simulator connected to five different ventilators. MFV allowed the use of lower inspiratory pressures and tidal volumes than conventional management of pressure control ventilation while maintaining adequate simulated gas exchange. Recently, a randomized controlled trial compared MFV to volume targeted ventilation (i.e., pressure control ventilation with adaptive targeting). Again, inspiratory pressures and tidal volumes were lower during midfrequency ventilation [29].
AVM2 was announced in 2017 [26] when van der Staay and Remus compared AVM2 with AVM and ASV using a lung simulator modeling a patient with restrictive lung disease. They demonstrated that minimizing inspiratory power (AVM2) results in higher frequencies, lower inspiratory pressure targets, and lower V_{T} compared with minimizing breathing power (AVM and ASV). The tidal volume dosage dropped from 7 to 5.3 mL/kg. However, there are currently no animal or human studies available with AVM2.
Results
Theoretical comparison of ASV, MFV, and AVM2
The optimal targeting schemes described above have one thing in common: for a given required alveolar minute ventilation and set of lung mechanics (for passive ventilation), they all suggest a ventilatory frequency that is optimal in some way (along with the associated optimal tidal volume). For a given alveolar minute ventilation, there are only two parameters which are free to vary (alveolar volume and frequency), and there are virtually infinite combinations. These targeting schemes offer three different ways to make the optimal selections. ASV is based on a model of unassisted breathing (sinusoidal pressure waveform driving function), under the assumption that the optimal tidal volume and frequency are those that would be picked by the patient’s brain to minimize power output of the muscles. AVM2 is based on a model of assisted breathing (square pressure waveform driving function) and selects tidal volume and frequency such that inspiratory power will be minimized to possibly avoid ventilatorinduced lung injury. Likewise, MFV was invented with the intention of serving lung protective ventilation by choosing the frequency that minimizes inspiratory pressure.
Currently, there is intensive research in the field of ventilationinduced lung injury, but definitive answers are still pending [31]. A further complication is the fact that the goals of mechanical ventilation (safety, comfort, and liberation) are often mutually exclusive. Hence, we did not seek to rate the performance of the optimum targeting schemes in terms of clinical outcomes because this is not possible without further experimental evidence. Instead, we determined how these targeting schemes behaved during selected simulation scenarios. This may promote understanding on a more intuitive level, instead of analyzing abstract mathematical equations.
Simulation parameters
We performed a comparison of the three targeting schemes by assuming four different mathematical simulation scenarios. The values of resistance and compliance were based on the work of Arnal et al. [32] for adult patients and of McCann et al. [33] for the neonatal scenario. With regard to dead space (V_{D}), there are several options. The physiological dead space, based on the Bohr equation (V_{DB}), is consistent with the equation V_{T} = V_{A} + V_{D}. [34]. ASV and AVM2 targeting schemes assume the dead space to be anatomical (V_{DA}), which is estimated to be 2.2 mL/kg [35] among all patient types regardless of the disease condition. In the ICU, V_{D}/V_{T} calculated with the blood gas measurement of PaCO_{2} is a useful indicator for the efficiency of ventilation. In this case, the dead space volume is calculated with the Enghoff modification of the Bohr equation (V_{DE}). Note that this volume does not necessarily exist physically and usually overestimates the physiological dead space volume (thus, it underestimates the required MV_{A}) [36]. To be clear, the presence of shunt and low \( \dot{V}/\dot{Q} \) are not dead space volumes but their effects manifest in the form of “virtual” dead space (the difference between V_{DE} and V_{DB}). Hence, if we want to simulate clinical experience using actual modes, we use V_{DA} for the frequency calculation of ASV and AVM2, respectively, and V_{DE} for MFV. However, we want to compare these targeting schemes at the same level of simulated PaCO_{2}. For this reason, we define the “Enghoff alveolar minute volume” as
which may underestimate the alveolar minute volume MV_{A} but is best correlated to PaCO_{2}. This reflects practical ventilation performance and efficiency as realistically as possible.
For the adult ARDS simulation, we assumed V_{DE} = 4.4 mL/kg [37], for normal adults V_{DE} = 2.2 mL/kg, and for adults with COPD V_{DE}/V_{T} = 0.49 [38] at V_{T}/kg = 8.9 mL/kg [32] which results also in V_{DE} = 4.4 mL/kg. For the neonate simulation, we assumed that a normal V_{DE} = 2.5 mL/kg [33], for RDS V_{DE} = 3.8 mL/kg [33], and for chronic lung disease (CLD) V_{DE} = 3.8 mL/kg [33].
The parameters of the lung models are listed in Table 2. Furthermore, we assumed passive inspiration and a linear, single compartment lung model with equal inspiratory and expiratory airway resistance. The ventilator inspiration to expiration ratio was assumed to be 1:1 because Otis’ model of breathing power is based on this ratio. Accordingly, it was ensured that the results are comparable among the different targeting schemes. With these assumptions, we calculated the optimum frequency with Eq. (13) for ASV, Eq. (18) for AVM2, and Eq. (23) for MFV and then derived tidal volume (V_{T} = MV_{A}/f + V_{D}), tidal pressure (V_{T}/C), tidal power (f·V_{T}·P_{T}/2), and Enghoff alveolar minute volume MV_{AE} by Eq. (25).
Calculation of the optimum frequency for the MFV target scheme was done experimentally using different simulated ventilator frequency setting values. We chose an interval of 0.1 breaths per minute to avoid excessive error. The same precision was applied for the iterative calculation of the ASV and AVM2 frequency. The simulation was done with the software package Matlab from Mathworks and comprises the following steps:

(a)
Set target alveolar minute volume (MV_{A})

(b)
Calculate and find the optimal frequency for ASV, AVM2, and MFV by the equations of the first section and the parameters defined above.

(c)
Calculate the resultant ventilation parameters of tidal volume, tidal pressure, tidal power, inspiratory power, and Enghoff alveolar minute volume for comparison.

(d)
Repeat steps (a) to (c) for different MV_{A} targets
Adults
For adult 70 kg simulation, we compare normal, ARDS, and COPD lung characteristics according to Table 2. Results are shown in Fig. 3. The green area highlights the range which is normally used for these patients according to Arnal et al. [32].
Neonatal
For neonatal simulation, we compare normal, RDS, and CLD lung characteristics according to Table 2. Results are shown in Fig. 4. The green areas highlight the range which is normally used for these patients according to McCann et al. [33].
Discussion
The effect of dead space
An attentive observer of Fig. 3 and Fig. 4 will notice that AVM2, which optimizes by minimizing inspiratory power, does not always have the lowest inspiratory power. On the other hand, MFV, which optimizes by minimizing inspiratory pressure, does not always have the smallest tidal pressure. These findings can be explained by the mismatch of dead space V_{D}. For ASV and AVM2, the calculation of the optimal frequencies is based on V_{DA} and not on V_{DE}. Therefore, if V_{DA} is significantly lower than V_{DE}, the Enghoff alveolar minute volume is underestimated. To get the same Enghoff alveolar minute volume, we increased the alveolar minute volume until the desired Enghoff alveolar minute volume was reached (resulting in the same simulated PaCO_{2} level). Hence, the calculation for the optimal frequencies for ASV, AVM2, and MFV cannot be done at the same alveolar minute volume to reach the same Enghoff alveolar minute volume. If V_{DE} is used for the optimal frequency calculation of ASV and AVM2, this mismatch will not occur, and AVM2 will have the lowest inspiratory power compared to ASV and MFV.
Given the assumption that V_{D} = 2.2 mL/kg, the initial settings using these targeting schemes will likely result in hypoventilation when actually ventilating patients with increased V_{D} (like ARDS and COPD). If, on the other hand, a factor based on measured physiologic V_{D} is used, then there is a greater likelihood that the initial result will be within the target range for CO_{2}. The main point of interest for clinicians relates to modes that require manual adjustment of the minute ventilation target (e.g., all conventional modes, ASV, and AVM1). For these modes, underestimating the required minute alveolar ventilation by use of an estimate of V_{D} that is too small when making initial ventilator settings (frequency and tidal volume) will result in subsequent manual adjustments and delay in achieving the PaCO_{2} target. For modes that automatically adjust the minute ventilation using ongoing monitoring of etPCO_{2}, subsequent adjustments are probably not as important an issue, but may still delay achievement of the desired PaCO_{2}.
Nevertheless, the simulation reflects the practical experience as realistically as possible because we have the same mismatch in reality. We believe that optimal targeting schemes might be more “optimal,” in terms of their optimization target, if they used V_{DB} instead of V_{DA} or V_{DE}. Otherwise, they have a mismatch between the model (V_{T} = V_{A} + V_{D}) and reality. On the other hand, there is clinical evidence that these targeting schemes work, with a few exceptions, for most lung conditions. If they are applied and understood correctly in practice, they may simplify clinical practice.
Optimal ventilation and VILI
In the first section of this work, we analyzed analytically how ventilation would look like if the clinical indicators tidal volume, tidal pressure, tidal power, or driving power were stringently optimized. For tidal volume and tidal pressure, the solution would be in the range of V_{T}~V_{D} at the highest possible frequency. At least for the ventilation of adults, this value seems to be far beyond the usual (green area in the figures). Also for the condition of minimal tidal power and driving power (V_{T} = 2·V_{D}), V_{T} seems too low for normal lungs. But this does not necessarily mean that they are bad predictors for VILI. On the other hand, we observe from Figs. 3 and 4 that the inspiratory power for the adult simulations is more than a decade higher than that of the neonatal scenarios. As Marini already observed, adjustment to the reduced baby lung capacity may be necessary. This indicates that the cost function, which describes optimal ventilation, does not have to be necessarily a good predictor for VILI. Therefore, the answer to the questions “What should we optimize?” and “What induces lung injury?” might be not the same. However, it seems reasonable to understand that a cost function describing optimal ventilation should approach, at least for patients at risk for lung injury, the cost function that describes VILI.
We see from the simulation that the results are supporting Marini’s proposal to adjust an indicator for VILI for compliance (as mentioned above). This concept also seems to be confirmed (at least in a mathematical sense) by the fact that tidal pressure (driving pressure) can be interpreted as a compliance adjusted tidal volume. Finally, the comparison between neonatal and adult scenarios suggests that tidal pressure is more related to the lung conditions than to the weight of the patient.
Limitations
For the simulations, we applied some simplifying assumptions and we did not include specific details reflecting the actual ventilator mode implementations of the targeting schemes. For example, ASV and AVM2 each have multiple “expert rules,” which could limit the frequency or tidal volume in certain scenarios (e.g., to avoid large tidal volumes or high intrinsic PEEP). Also, ASV is currently not designed for neonatal ventilation and limits its frequency to between 5 and 60 breaths per minute. Furthermore, AVM2 actually assumes an I:E ratio of 1:1.8, not 1:1, which was empirically derived during design of the mode implementation. This would lead to higher tidal volumes for normal lung conditions. MFV is designed to allow accommodation of unequal inspiratory and expiratory airway resistances.
Conclusions
Modes of mechanical ventilation have shown a steady evolution over the last four decades. They have increased in complexity as engineers attempt to add technical capabilities that better serve clinical goals. A key feature of this complexity is the development of new targeting schemes, moving away from simple setpoint targeting (all targets are operator preset) through adaptive targeting (some targets are automatically adjusted) to optimal targeting (targets are automatically adjusted to maximize or minimize some desired performance characteristic) and even intelligent targeting (automatic adjustment and selection of targets using the tools of artificial intelligence). In particular, optimum targeting schemes have been the central feedback control mechanisms of the most complex modes currently available. Optimization means, by definition, that there exists no better alternative to get, do, or set something, given the constraints of the mathematical model used. But we have shown that this kind of targeting scheme for ventilator modes is based on fairly arbitrary assumptions and presupposes clearly defined goals and targets, which are still topics of clinical debate. This paper tries to clarify these assumptions and point out that thinking about what should be optimized is much more important than thinking about how we should optimize. We suggest that optimization based on tidal volume, tidal pressure, or tidal power as the sole criteria may result in unusual ventilation strategies and settings. Therefore, we can adumbrate that modes based only on one of these variables may have limited clinical success.
Additional information
All authors have approved this submission. The content of this manuscript has not been published, or submitted for publication elsewhere.
Abbreviations
 \( \widehat{\dot{V}} \) :

Peak flow
 \( {\dot{W}}_{\mathrm{E}} \) :

Elastic power
 \( {\dot{W}}_{\mathrm{insp}} \) :

Inspiratory power
 \( {\dot{W}}_{\mathrm{mus}} \) :

Breathing power (muscular power)
 \( {\dot{W}}_{\mathrm{PEEP}} \) :

PEEP power
 \( {\dot{W}}_{\mathrm{R}} \) :

Resistive power
 \( {\dot{W}}_{\mathrm{T}} \) :

Tidal power
 \( \dot{V} \) :

Flow
 ASV:

Adaptive support ventilation
 AVM:

Adaptive ventilation mode
 C:

Compliance
 D:

Fraction of the inspiration time Ti to the period T
 f :

Frequency (in Hz)
 f _{BF} :

Frequency at minimal breathing force
 f _{BP} :

Frequency at minimal breathing power
 f_{IP} :

Frequency at minimal inspiratory power
 f_{TP} :

Frequency at minimal tidal power
 MFV:

MIDfrequency ventilation
 MV:

Minute volume measured at the proximal airway
 MV_{A} :

Alveolar minute volume
 MV_{AE} :

Alveolar minute volume, calculated with the Enghoff modification of the Bohr equation
 P _{insp} :

Inspiratory pressure
 P _{mus} :

Pressure generated by inspiratory muscles
 P _{plat} :

Plateau pressure
 P _{T} :

Tidal pressure (aka driving pressure)
 R :

Airway resistance (linear)
 R′:

Airway resistance (nonlinear portion)
 RC:

Time constant (product of resistance and compliance)
 V :

Volume
 V _{D} :

Dead space volume
 V _{DA} :

Anatomical dead space
 V _{DB} :

Physiological dead space, calculated with the Bohr equation
 V _{DE} :

Physiological dead space, calculated with the Enghoff modification of the Bohr equation
 V _{T} :

Tidal volume
References
 1.
Chatburn RL, ElKhatib M, MirelesCabodevila E (2014) A taxonomy for mechanical ventilation: 10 fundamental maxims. Respir Care 59(11):1747–1763
 2.
Brower RG et al (2000) Ventilation with lower tidal volumes as compared with traditional tidal volumes for acute lung injury and the acute respiratory distress syndrome. N Engl J Med 342(18):1301–1308
 3.
Bowton DL, Scott LK (2016) Ventilatory management of the noninjured lung. Clin Chest Med 37(4):701–710
 4.
Serpa Neto A, Nagtzaam L, Schultz MJ (2014) Ventilation with lower tidal volumes for critically ill patients without the acute respiratory distress syndrome: a systematic translational review and metaanalysis. Curr Opin Crit Care 20(1):25–32
 5.
Neto AS, Simonis FD, Barbas CS et al (2015) Lungprotective ventilation with low tidal volumes and the occurrence of pulmonary complications in patients without acute respiratory distress syndrome: a systematic review and individual patient data analysis. Crit Care Med 43(10):2155–2163
 6.
Serpa Neto A, Simonis FD, Schultz MJ (2015) How to ventilate patients without acute respiratory distress syndrome? Curr Opin Crit Care 21(1):65–73
 7.
Brochard L, Slutsky A, Pesenti A (2017) Mechanical ventilation to minimize progression of lung injury in acute respiratory failure. Am J Respir Crit Care Med 195(4):438–442
 8.
Mauri T, Yoshida T, Bellani G et al (2016) Esophageal and transpulmonary pressure in the clinical setting: meaning, usefulness and perspectives. Intensive Care Med 42(9):1360–1373
 9.
Borges JB, Hedenstierna G, Larsson A, SuarezSipmann F (2015) Altering the mechanical scenario to decrease the driving pressure. Crit Care 19(1):342
 10.
Amato MBP, Meade MO, Slutsky AS et al (2015) Driving pressure and survival in the acute respiratory distress syndrome. N Engl J Med 372(8):747–755
 11.
Tejerina E, Pelosi P, Muriel A et al (2017) Association between ventilatory settings and development of acute respiratory distress syndrome in mechanically ventilated patients due to brain injury. J Crit Care 38:341–345
 12.
Serpa Neto A, Schmidt M, Azevedo LCP et al (2016) Associations between ventilator settings during extracorporeal membrane oxygenation for refractory hypoxemia and outcome in patients with acute respiratory distress syndrome: a pooled individual patient data analysis: mechanical ventilation during ECMO. Intensive Care Med 42(11):1672–1684
 13.
Neto AS, Hemmes SNT, Barbas CSV et al (2016) Association between driving pressure and development of postoperative pulmonary complications in patients undergoing mechanical ventilation for general anaesthesia: a metaanalysis of individual patient data. Lancet Respir Med 4(4):272–280
 14.
Guérin C, Papazian L, Reignier J et al (2016) Effect of driving pressure on mortality in ARDS patients during lung protective mechanical ventilation in two randomized controlled trials. Crit Care 20(1):384
 15.
Gattinoni L, Tonetti T, Cressoni M et al (2016) Ventilatorrelated causes of lung injury: the mechanical power. Intensive Care Med 42(10):1567–1575
 16.
Marini JJ, Jaber S (2016) Dynamic predictors of VILI risk: beyond the driving pressure. Intensive Care Med 42(10):1597–1600
 17.
Cressoni M, Gotti M, Chiurazzi C, Massari D, Algieri I, Amini M, Cammaroto A, Brioni M, Montaruli C, Nikolla K, Guanziroli M, Dondossola D, Gatti S, Valerio V, Vergani GL, Pugni P, Cadringher P, Gagliano NGL, Gattinoni L, Gagliano N et al (2016) Mechanical power and development of ventilatorinduced lung injury. Anesthesiology 124(5):1100–1108
 18.
Otis AB, Fenn WO, Rahn H (1950) Mechanics of breathing in man. J Appl Physiol 2(11):592–607
 19.
Mead J (1960) Control of respiratory frequency. J Appl Physiol 15(3):325–336
 20.
Tehrani FT (2008) Automatic control of mechanical ventilation. Part 1: theory and history of the technology. J Clin Monit Comput 22(6):409–415
 21.
Laubscher TP, Frutiger A, Fanconi S, Brunner JX (1996) The automatic selection of ventilation parameters during the initial phase of mechanical ventilation. Intensive Care Med 22(3):199–207
 22.
Laubscher TP, Frutiger A, Fanconi S et al (1994) Automatic selection of tidal volume, respiratory frequency and minute ventilation in intubated ICU patients as startup procedure for closedloop controlled ventilation. Int J Clin Monit Comput 11(1):19–30
 23.
Dongelmans DA, Veelo DP, Bindels A et al (2008) Determinants of tidal volumes with adaptive support ventilation: a multicenter observational study. Anesth Analg 107(3):932–937
 24.
Sulemanji D, Kacmarek R (2010) Adaptive support ventilation: an inappropriate mechanical ventilation strategy for acute respiratory distress syndrome? Anesthesiology 111(5):1295–1296
 25.
Arnal JM, Garnero A, Novonti D et al (2013) Feasibility study on full closedloop control ventilation (IntelliVentASV™) in ICU patients with acute respiratory failure: a prospective observational comparative study. Crit Care 17(5):R196
 26.
van der Staay M, Remus C (2017) Adaptive ventilation mode 2. https://downloads.imt.ch/usdavkarsv/scientificNote_AVM2.pdf. Accessed 12 Mar 2018
 27.
Marini JJ, Crooke PS 3rd (1993) A general mathematical model for respiratory dynamics relevant to the clinical setting. Am Rev Respir Dis 147(1):14–24
 28.
MirelesCabodevila E, Chatburn RL (2008) Original contributions midfrequency ventilation: unconventional use of conventional mechanical ventilation as a lungprotection strategy. Respir Care 53(12):1669–1677
 29.
Bhat R, Kelleher J, Ambalavanan N et al (2017) Feasibility of midfrequency ventilation among infants with respiratory distress syndrome. Respir Care 62(4):481–488
 30.
Hewlett AM, Platt AS, Terry VG (1977) Mandatory minute volume. A new concept in weaning from mechanical ventilation. Anaesthesia 32(2):163–169
 31.
Marini JJ (2018) Dissipation of energy during the respiratory cycle: conditional importance of ergotrauma to structural lung damage. Curr Opin Crit Care 24(1):16–22
 32.
Arnal JM, Garnero A, Saoli M, Chatburn RL (2018) Parameters for simulation of adult patients during mechanical ventilation. Respir Care 63(2):158 LP–158168
 33.
McCann EM, Goldman SL, Brady JP (1987) Pulmonary function in the sick newborn infant. Pediatr Res 21(4):313–325
 34.
SuarezSipmann F, Bohm SH, Tusman G (2014) Volumetric capnography: the time has come. Curr Opin Crit Care 20(3):333–339
 35.
Radford EP, Ferris BG, Kriete BC (1954) Clinical use of a nomogram to estimate proper ventilation during artificial respiration. N Engl J Med 251(22):877–884
 36.
Tusman G, Sipmann FS, Bohm SH (2012) Rationale of dead space measurement by volumetric capnography. Anesth Analg 114(4):866–874
 37.
Kallet RH, Zhuo H, Ho K et al (2017) Lung injury etiology and other factors influencing the relationship between deadspace fraction and mortality in ARDS. Respir Care 62(10):1241–1248
 38.
Farah R, Makhoul N (2009) Can dead space fraction predict the length of mechanical ventilation in exacerbated COPD patients? Int J Chron Obstruct Pulmon Dis 4:437–441
Author information
Affiliations
Contributions
Both authors participated in writing and review of the manuscript. Both authors read and approved the final manuscript.
Corresponding author
Correspondence to Matthias van der Staay.
Ethics declarations
Ethics approval and consent to participate
Not applicable
Consent for publication
Not applicable
Competing interests
Robert Chatburn is a consultant for imtmedical AG, Switzerland; IngMar Medical Inc., Pittsburgh, USA; and Drive DeVilbiss Healthcare, Port Washington, NY, USA. Matthias van der Staay is an employee of IMT AG and works for imtmedical AG.
Publisher’s Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Additional file
Additional file 1:
Supplementary material. (DOCX 1009 kb)
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
About this article
Cite this article
van der Staay, M., Chatburn, R.L. Advanced modes of mechanical ventilation and optimal targeting schemes. ICMx 6, 30 (2018) doi:10.1186/s4063501801950
Received
Accepted
Published
DOI
Keywords
 Mechanical ventilation
 Mathematical modeling
 Lung protective ventilation
 Optimal targeting schemes
 Simulation