 Research
 Open access
 Published:
Bias flow rate and ventilation efficiency during adult highfrequency oscillatory ventilation: a lung model study
Intensive Care Medicine Experimental volume 6, Article number: 11 (2018)
Abstract
Background
Bias flow (BF) is essential to maintain mean airway pressure (MAP) and to washout carbon dioxide (CO_{2}) from the oscillator circuit during highfrequency oscillatory ventilation (HFOV). If the BF rate is inadequate, substantial CO_{2} rebreathing could occur and ventilation efficiency could worsen. With lower ventilation efficiency, the required stroke volume (SV) would increase in order to obtain the same alveolar ventilation with constant frequency. The aim of this study was to assess the effect of BF rate on ventilation efficiency during adult HFOV.
Methods
The R100 oscillator (Metran, Japan) was connected to an original lung model internally equipped with a simulated bronchial tree. The actual SV was measured with a flow sensor placed at the Ypiece. Carbon dioxide (CO_{2}) was continuously insufflated into the lung model (\( \dot{\mathrm{V}} \)CO_{2}), and the partial pressure of CO_{2} (PCO_{2}) in the lung model was monitored. Alveolar ventilation (\( \dot{\mathrm{V}} \)A) was estimated as \( \dot{\mathrm{V}} \)CO_{2} divided by the stabilized value of PCO_{2}. \( \dot{\mathrm{V}} \)A was evaluated by setting SV from 80 to 180 mL (10 mL increments, n = 5) at a frequency of 8 Hz, a MAP of 25 cmH_{2}O, and a BF of 10, 20, 30, and 40 L/min (study 1). Ventilation efficiency was calculated as \( \dot{\mathrm{V}} \)A divided by the actual minute volume. The experiment was also performed with an actual SV of 80, 100, and 120 mL and a BF from 10 to 60 L/min (10 L/min increments: study 2).
Results
Study 1: With the same setting SV, the \( \dot{\mathrm{V}} \)A with a BF of 20 L/min or more was significantly higher than that with a BF of 10 L/min. Study 2: With the same actual SV, the \( \dot{\mathrm{V}} \)A and the ventilation efficiency with a BF of 30 L/min or more were significantly higher than those with a BF of 10 or 20 L/min.
Conclusions
Increasing BF up to 30 L/min or more improved ventilation efficiency in the R100 oscillator.
Background
Bias flow (BF) is essential to maintain mean airway pressure (MAP) and to washout carbon dioxide (CO_{2}) from the oscillator circuit (“CO_{2} washout”) during highfrequency oscillatory ventilation (HFOV). If the BF rate is inadequate, substantial CO_{2} rebreathing could occur, and the resultant hypercapnia might become a problem [1, 2]. This situation increases wasted ventilation and worsens ventilation efficiency. With lower ventilation efficiency, the required stroke volume (SV) must increase in order to obtain the same alveolar ventilation with constant frequency.
The SensorMedics 3100B oscillator (CareFusion, Yorba Linda, CA, USA) has often been used with a BF of 30 or 40 L/min [3,4,5]. For the R100 oscillator (Metran Co. Ltd., Kawaguchi, Saitama, Japan), a BF of 20 to 40 L/min is recommended in Japan, although a BF of 30 L/min or more has been preferred in many institutions, especially in patients with spontaneous breathing. Some studies have investigated the effect of the BF rate [1, 2, 6, 7], although none have reported the exact effect of the BF rate on ventilation efficiency during adult HFOV. The aim of this study is to assess the effect of the BF rate on ventilation efficiency using the R100 oscillator.
Methods
Experimental setting
The R100 oscillator was connected to the lung model via an angletype connector and an endotracheal tube (ETT) with an internal diameter of 8.0 mm and a length of 30 cm (Fig. 1). A microelectromechanical systems mass flow sensor (Siargo FS6022B150, Siargo Ltd., Santa Clara, CA, USA) was placed between the angletype connector and the Ypiece for the actual SV (aSV) measurement. The total dead space volume (VD) was approximately 110 mL, and the airway resistance was approximately 2.0 cmH_{2}O/L/s. The common oscillator settings during the experiments were as follows: frequency of 8 Hz, MAP of 25 cmH_{2}O, and fraction of inspired oxygen of 0.21. The inspiratory time was fixed at 50% in this oscillator. The heated humidifier was turned off.
Measurement of actual stroke volume (aSV)
The aSV was measured with a prototype SV measurement system (Metran Co. Ltd., Kawaguchi, Saitama, Japan). In this system, the analogue flow signal was sampled at 200 Hz and digitally integrated to determine expiratory SV every second using a computer data acquisition system (LabView Ver. 14, National Instruments, Austin, TX, USA). The mean value of 60 data measurements for 1 min was calculated as the aSV.
Lung model
The lung model consisted of a 20L airtight rigid plastic container internally equipped with a simulated bronchial tree (KYOTO KAGAKU Co. Ltd., Kyoto, Japan) that had three to seven steps of bifurcations to 20 segmental bronchial branches (Fig. 1). The top of the ETT was located 3.5 cm from the carina. The total 20 L volume of the container accounts for an adiabatic static compliance of 19.3 mL/cmH_{2}O (approximately equal to severe ARDS [8]) due to gas compression. The container had two ports for gas insufflation and for gas sampling. A small electrical fan was placed horizontally on the bottom to assist gas mixing in the container.
Measurement of alveolar ventilation (\( \dot{\mathrm{V}} \)a)
CO_{2} was insufflated into the lung model at approximately 200 mL/min (floattype area flowmeter), and continuous gas sampling was performed from the lung model using a capnometer (Life Scope TR, NIHON KOHDEN Co., Tokyo, Japan) (Fig. 1). The gas sampling rate of the capnometer was set at 200 mL/min. The partial pressure of CO_{2} (PCO_{2}; mmHg) was monitored by the capnometer, and the stabilized value was recorded. The actual minute volume of insufflated CO_{2} (\( \dot{\mathrm{V}} \)CO_{2}; mL/min) was calculated by the measurement of PCO_{2} using the capnometer when it was mixed with oxygen at 5 L/min (fixedtype flowmeter). This was performed before starting the experiment and was confirmed every 1 to 2 h. The \( \dot{\mathrm{V}} \)A was estimated by applying the alveolar ventilation equation (PCO_{2} = 0.863 × \( \dot{\mathrm{V}} \)CO_{2}/\( \dot{\mathrm{V}} \)A, PCO_{2} and \( \dot{\mathrm{V}} \)A: BTPS, \( \dot{\mathrm{V}} \)CO_{2}: STPD), though the equation was rearranged and used as \( \dot{\mathrm{V}} \)A = \( \dot{\mathrm{V}} \)CO_{2}/PCO_{2} because all experiments were done under room temperature and dry conditions.
Although the SV can be set at up to 205 mL with a frequency of 8 Hz in the R100 oscillator, the measurable range of the flow sensor was limited. Therefore, \( \dot{\mathrm{V}} \)A was evaluated with the SV setting (sSV) from 80 to 180 mL (10 mL increments) and with the BF from 10 to 40 L/min (10 L/min increments; study 1). Additionally, the ratio of \( \dot{\mathrm{V}} \)A to the actual minute ventilation volume (a\( \dot{\mathrm{V}} \)E; aSV/1000 × 8 × 60, L/min) was calculated (\( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E) as an index for ventilation efficiency. \( \dot{\mathrm{V}} \)A was also evaluated with the targeted aSV of 80, 100, and 120 mL and with the BF of 10 to 60 L/min (10 L/min increments; study 2). Because the BF rate can be set from 10 to 40 L/min in the R100 oscillator, air was added to the BF supply port using a floattype area flowmeter. To further examine the role of BF, the relationship between \( \dot{\mathrm{V}} \)A and a\( \dot{\mathrm{V}} \)E/BF was investigated in study 2.
Each experiment was conducted five times.
Statistical analysis
The statistical analysis was performed with BellCurve for Excel ver. 2.02 (SSRI Co. Ltd., Tokyo, Japan) using oneway analysis of variance followed by Tukey’s test. P < 0.05 was considered statistically significant. Curve fitting was also performed by the same software.
Results
Study 1
Figure 2 shows the relationships between sSV and aSV at respective BF rates. The aSV was proportional to the sSV at all BF rates. Increasing the BF rate decreased the aSV at most settings. In all sSVs, the aSV with a BF of 30 or 40 L/min was significantly lower than that with a BF of 10 or 20 L/min (P < 0.001). The airway pressure amplitude (AMP) measured at the Ypiece (displayed on the panel of the R100 oscillator) showed a similar change (Additional file 1). Figure 3 shows the relationships between sSV and \( \dot{\mathrm{V}} \)A at respective BF rates. The \( \dot{\mathrm{V}} \)A was correlated to the power of sSV at all BF rates. The perfect powers were 1.226, 1.580, 1.744, and 1.819 with BFs of 10, 20, 30, and 40 L/min, respectively. In all sSVs, the \( \dot{\mathrm{V}} \)A with a BF of 20 L/min or more was significantly higher than the \( \dot{\mathrm{V}} \)A with a BF of 10 L/min (the P value was different for the sSV). The \( \dot{\mathrm{V}} \)A with a BF of 30 or 40 L/min was not significantly higher than the \( \dot{\mathrm{V}} \)A with a BF of 20 L/min in all sSVs. The \( \dot{\mathrm{V}} \)A with BFs of 30 and 40 L/min was not significantly different in all sSVs.
Figure 4 shows the relationships between aSV and \( \dot{\mathrm{V}} \)A at respective BF rates (individual data). The \( \dot{\mathrm{V}} \)A was correlated with the power of the aSV at all BF rates. The perfect power were 1.128, 1.528, 1.631, and 1.664 with BFs of 10, 20, 30, and 40 L/min, respectively. It appeared that the \( \dot{\mathrm{V}} \)A increased with an increasing BF up to 30 L/min with an aSV of more than approximately 100 mL (a statistical analysis was not conducted).
Figure 5 shows the relationships between aSV and \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E at respective BF rates (individual data). The \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E was proportional to the aSV at all BF rates. It appeared that the \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E increased with an increasing BF up to 30 L/min with an aSV of more than approximately 100 mL (a statistical analysis was not conducted).
Study 2
aSVs with the target of 80, 100, and 120 mL were not significantly different at respective BF rates (Additional file 2). Figure 6 shows the \( \dot{\mathrm{V}} \)A, and Fig. 7 shows the \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E with the targeted aSV of 80, 100, and 120 mL. In all targeted aSVs, the \( \dot{\mathrm{V}} \)A and the \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E with a BF of 20 L/min or more were significantly higher than those with a BF of 10 L/min (P < 0.001), and those with a BF of 30 L/min or more were significantly higher than those with a BF of 20 L/min (P < 0.001). In all targeted aSVs, the \( \dot{\mathrm{V}} \)A and the \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E with a BF of 50 or 60 L/min were significantly higher than those with a BF of 30 L/min (the P value was different for the targeted aSV with a BF of 50 L/min, and P < 0.001 with a BF of 60 L/min).
Figure 8 shows the relationships between a\( \dot{\mathrm{V}} \)E/BF and \( \dot{\mathrm{V}} \)A with respective targeted aSVs (individual data). The \( \dot{\mathrm{V}} \)A was correlated to the exponential of the a\( \dot{\mathrm{V}} \)E/BF with all targeted aSVs. The Yintercepts were 7.54 with a targeted aSV of 80 mL, 11.46 with a targeted aSV of 100 mL, and 15.31 with a targeted aSV of 120 mL.
Discussion
We investigated the effect of the BF rate on alveolar ventilation and ventilation efficiency using an original lung model in the adult oscillator R100, and our study showed that increasing the BF rate improved the ventilation efficiency up to 30 L/min or more with a frequency of 8 Hz. This result is almost consistent with the previous large animal study [1], though the used frequency, oscillator, and the structure of the circuit were different from our study. In Japan, the R100 oscillator has been often used with the frequency around 8 Hz or more. In the OSCAR trial which used the R100 oscillator, the mean frequency was 7.8 Hz on the first day [9]. We therefore performed the experiments with a frequency of 8 Hz. The initial setting of the BF rate was 20 L/min, and there was no description of BF change in the management protocol in the OSCAR trial [9]. If the BF rate is inadequate, the resultant higher aSV might tend to impair the benefit of HFOV. In the animal study which showed the superiority of higher frequency for lung protection, the aSV was lower with higher frequency [10]. In their study, it was unclear whether the higher frequency or the lower aSV was more important. However, it might be conceivable that the lower aSV might be beneficial in HFOV as well as in conventional lung protective ventilation [11]. The improvement of ventilation efficiency might be a future direction for reaching the goal of HFOV.
In the previous studies that measured aSV with the 3100B oscillator, a pneumotachometer or a hotwire anemometer was used with a data sampling rate of 1000 Hz [12, 13]. Iguchi et al. measured aSV with two adult oscillators (3100B and R100) using a position sensor placed on the lung model with a data sampling rate of 667 Hz [14]. Although the aSV measurements were steady and stable in our study, the data sampling rate of 200 Hz might not be adequate for the frequency of 8 Hz. Therefore, we averaged the 60 data points for 1 min. However, a higher sampling rate would be desirable for robust reliability of the aSV measurement.
We have used a 20L plastic container as a lung model [15, 16], and we improved it by adding a simulated bronchial tree in this study. An advantage of a lung model study is that a wide range of sSVs can be evaluated. However, the simulated bronchial tree in our lung model was quite different from a real lung, with the airway resistance being low and the VD was rather small for an adult. Therefore, the obtained values of \( \dot{\mathrm{V}} \)A and \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E cannot be applied in clinical situations, and the interpretation of the current results may be difficult. For example, the \( \dot{\mathrm{V}} \)A would be approximately 21 L/min with a frequency of 8 Hz, a sSV of 205 mL (maximum), and a BF of 20 L/min in Fig. 3. These settings might be almost equal to the mean setting values of the first day in the OSCAR trial (7.8 Hz and 213 mL) [9]. On the other hand, because the phenomenon that occurs in the lung model must be almost identical with the same aSV and the same frequency, the observed effect of the BF rate in our study must have been caused by different phenomena that occurred in the oscillator circuit. Therefore, it is conceivable that the effect of the BF rate on the ventilation efficiency has a similar tendency in clinical situations.
In clinical or experimental settings, the \( \dot{\mathrm{V}} \)A during HFOV had been determined as frequency^{a} × SV^{b} (the values for a and b were approximately 1 and 2, respectively) [17]. Therefore, we applied the power approximation curves to the relationships between sSV or aSV and \( \dot{\mathrm{V}} \)A, and those were well fitted (Figs. 3 and 4). This finding would indicate that our lung model reproduced the particular ventilation mechanism of HFOV to some extent. Although the number of the power was less than two possibly because of the incomplete bronchial tree model, the number of the power was affected by the BF rate (Figs. 3 and 4). We used \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E as an index of ventilation efficiency in the same way as conventional ventilation, and it was proportional to the aSV (Fig. 5). However, increasing aSV did not increase the ventilation efficiency with the BF of 10 L/min (Figs. 5 and 7). Theoretically, the \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E must be very low, especially in the range of aSV far less than VD, and it must increase with higher aSV, especially in the range of aSV far more than VD, because of the correlation of direct alveolar ventilation (i.e., convection). However, “CO_{2} washout” from the oscillator circuit would be considerably involved in this issue as discussed in the next paragraph.
In the adult oscillator, the oscillation unit is located on the inspiratory circuit side. Therefore, the major part of the exhaled gas would be pulled out to the inspiratory circuit during the expiratory phase of the oscillation. This has been described as “retrograde CO_{2} entrainment” in the 3100B oscillator [2], and some CO_{2} rebreathing could naturally occur [1]. Although some gas regurgitation from the expiratory circuit to the inspiratory circuit could occur during the expiratory phase of the oscillation, major gas regurgitation would be prevented by the oneway valve placed at the end of the expiratory circuit in the R100 oscillator. The exhaled gas would mix with the fresh gas in the inspiratory circuit, and the CO_{2} would be diluted. Then, the mixed gas would go to the lung (aSV) and to the expiratory circuit (BF/cycle) during the next inspiratory phase of the oscillation. The former would include CO_{2} rebreathing, and the latter would be “CO_{2} washout.” The dilution of exhaled CO_{2} by the gas mixing in the inspiratory circuit would basically be affected by the ratio of the aSV and the BF rate per cycle (BF/cycle), that is, the ratio of the a\( \dot{\mathrm{V}} \)E and the BF rate per minute. Based on this concept, we examined the relationship between \( \dot{\mathrm{V}} \)A and a\( \dot{\mathrm{V}} \)E/BF (Fig. 8). Exponential curve fitting was used so that the Xintercept would be infinite because the \( \dot{\mathrm{V}} \)A would be zero without the BF. Although we performed the experiments with a single frequency (8 Hz), Fig. 8 would suggest that increasing the aSV or increasing the frequency would increase the need for a BF rate. The former could be a factor that affected the relationship between the aSV and \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E (Fig. 5), as discussed in the paragraph above. Increasing aSV would promote the ventilation phenomena occurring in the lung model, although the need for BF would increase. Then, increasing aSV would induce a relative insufficiency of BF because of the constant BF rate in our study. In the case with the BF of 10 L/min, a\( \dot{\mathrm{V}} \)E/BF is 3.84, 4.8, and 5.76 with the aSV of 80, 100, and 120 mL, respectively. To keep an a\( \dot{\mathrm{V}} \)E/BF of 3.84, BF rates of 12.5 and 15 L/min are necessary with the aSV of 100 and 120 mL, respectively. This would be a reason why increasing aSV did not increase the ventilation efficiency with the BF of 10 L/min (Figs. 5 and 7). All results of our study could be influenced by this issue and must be carefully interpreted. Turner et al. reported that increasing the BF rate had little effect on the ventilation using an infant oscillator (3100A) with BF rates of 10, 20, 30, and 40 L/min [7], although we must consider the fact that they used a juvenile swine (mean body weight 15.1 kg) and a relatively low aSV would be used. On the other hand, because it is conceivable that CO_{2} rebreathing does not occur with an infinite BF rate, the Yintercepts in Fig. 8 would indicate the upper limits of the improvement of \( \dot{\mathrm{V}} \)A by increasing the BF rate.
The gas mixing that occurred in the inspiratory circuit could not be completed. Conceptually, the BF might be separated from the gas that completely mixes with the exhaled gas (effective BF) and the gas that does not mix with the exhaled gas (ineffective BF). The mean concentration of CO_{2} exhausted from the expiratory circuit in the steady state must be \( \dot{\mathrm{V}} \)CO_{2}/BF. In the case that all BFs are effective BFs, the mean concentration of rebreathing CO_{2} must be equal to this value. Namely, mean rebreathing PCO_{2} would be 15.2, 7.6, 5.1, and 3.8 mmHg with BFs of 10, 20, 30, and 40 L/min, respectively, when the \( \dot{\mathrm{V}} \)CO_{2} is 200 mL/min. If some BFs are ineffective BFs, rebreathing PCO_{2} might be higher. Although the ratio of effective BF is unclear, this concept may help us to understand the mechanism of “CO_{2} washout” by the BF.
Conclusions
In this lung model study, ventilation efficiency improved at a BF equal to or more than 30 L/min compared to a BF equal to or less than 20 L/min with the R100 oscillator. The mechanism of improving ventilation efficiency by increasing the BF rate would be the decreased CO_{2} rebreathing, and the need for a BF rate would depend on a\( \dot{\mathrm{V}} \)E.
Abbreviations
 \( \dot{\mathrm{V}} \)A:

Alveolar ventilation
 \( \dot{\mathrm{V}} \)A/a\( \dot{\mathrm{V}} \)E:

Ratio of alveolar ventilation to actual minute ventilation
 a\( \dot{\mathrm{V}} \)E:

Actual minute ventilation
 a\( \dot{\mathrm{V}} \)E/BF:

Ratio of actual minute ventilation to bias flow rate
 AMP:

Airway pressure amplitude
 ARDS:

Acute respiratory distress syndrome
 aSV:

Actual stroke volume
 BF:

Bias flow
 \( \dot{\mathrm{V}} \)CO_{2} :

Minute volume of insufflated CO_{2}
 CO_{2} :

Carbon dioxide
 ETT:

Endotracheal tube
 HFOV:

Highfrequency oscillatory ventilation
 MAP:

Mean airway pressure
 PCO_{2} :

Partial pressure of CO_{2}
 RCT:

Randomized controlled trial
 sSV:

Setting stroke volume
 SV:

Stroke volume
 VD:

Dead space volume
References
Lunkenheimer PP, Redmann K, Stroh N, Gleich C, Krebs S, Scheld HH, Dietl KH, Fischer S, Whimster WF (1994) Highfrequency oscillation in an adult porcine model. Crit Care Med 22(9 Suppl):S37–S48
Bostick AW, Naworol GA, Britton TJ, Ori TR, French SK, Derdak S (2012) Inspiratory limb carbon dioxide entrainment during highfrequency oscillatory ventilation; characterization in a mechanical test lung and swine model. Respir Care 57(11):1865–1872
Fort P, Farmer C, Westeman J, Johannigman J, Beninati W, Dolan S, Derdak S (1997) Highfrequency oscillatory ventilation for adult respiratory distress syndrome—a pilot study. Crit Care Med 25(6):937–947
Derdak S, Mehta S, Stewart T, Rogers M, Buchman TG, Carlin B, Lowson S, and the Multicenter Oscillatory Ventilation for Acute Respiratory Distress Syndrome Trial (MOAT) Study Investigators (2002) Highfrequency oscillatory ventilation for acute respiratory distress syndrome in adults. A randomized, controlled trial. Am J Respir Crit Care Med 166(6):801–808
Fessler HE, Derdak S, Ferguson ND, Hager DN, Kacmarek RM, Thompson T, Brower RG (2007) A protocol for highfrequency oscillatory ventilation in adults: results from a roundtable discussion. Crit Care Med 35(7):1649–1654
Solway J, Gaveriely N, Slutsky AS, Rossing TH, Drinker P, Saari AF, Drazen JM (1985) Effect of bias flow rate on gas transport during highfrequency oscillatory ventilation. Respir Physiol 60(2):267–276
Turner DA, Adams DF, Gentile MA, Lee W, Quick GA, Smith B, Cheifetz IM (2012) Bias flow does not affect ventilation during high frequency oscillatory ventilation in a pediatric animal model of acute lung injury. Pediatric Crit Care Med 13(2):e108–e112
Murray JF, Matthay MA, Luce JM, Flick MR (1988) An expanded definition of the adult respiratory distress syndrome. Am Rev Respir Dis 138(3):720–723
Young D, Lamb SE, Shah S, MacKenzie I, Tunnicliffe W, Lall R, Rowan K, Cuthbertson BH, OSCAR Study Group (2013) Highfrequency oscillation for acute respiratory distress syndrome. New Engl J Med 368(9):806–813
Liu S, Yi Y, Wang M, Chen Q, Huang Y, Liu L, Xie J, Zhou D, Qiu H (2013) Higher frequency ventilation attenuates lung injury during highfrequency oscillatory ventilation in sheep models of acute respiratory distress syndrome. Anesthesiology 119(2):398–411
Acute Respiratory Distress Syndrome Network, Brower RG, Matthay MA, Morris A, Schoenfeld D, Thompson BT, Wheeler A (2000) Ventilation with lower tidal volumes as compared with traditional tidal volumes for acute lung injury and the acute respiratory distress syndrome. N Engl J Med 342(18):1301–1308
Sedeek KA, Takeuchi M, Suchodolski K, Kacmarek RM (2003) Determinants of tidal volume during highfrequency oscillation. Crit Care Med 31(1):227–231
Hager DN, Fessler HE, Kaczka DW, Shanholtz CB, Fuld MK, Simon BA, Brower RG (2007) Tidal volume delivery during highoscillatory ventilation in adults with acute respiratory distress syndrome. Crit Care Med 35(6):1522–1529
Iguchi N, Hirao O, Uchiyama A, Mashimo T, Nishimura M, Fujino Y (2010) Evaluation of performance of two highfrequency oscillatory ventilators using a model lung with a position sensor. J Anesth 24(6):888–892
Shiba N, Nagano O, Hirayama T, Ichiba S, Ujike Y (2012) Humidification of base flow gas during adult highfrequency oscillatory ventilation: an experimental study using a lung model. Acta Med Okayama 66(4):335–341
Hirayama T, Nagano O, Shiba N, Yumoto T, Sato K, Terado M, Ugawa T, Ichiba S, Ujike Y (2014) Mean lung pressure during adult highfrequency oscillatory ventilation: an experimental study using a lung model. Acta Med Okayama 68(6):323–329
Pillow JJ (2005) Highfrequency oscillatory ventilation: mechanisms of gas exchange and lung mechanics. Crit Care Med 33(3):135–141
Acknowledgements
The authors thank the Metran Co. Ltd. for the use of the prototype SV measurement system.
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the corresponding author on reasonable request.
Author information
Authors and Affiliations
Contributions
The study was designed by ON and TY. ON, TY, AN, SK, TF, and SA collected and analyzed the data. ON and TY prepared the manuscript. HY reviewed the manuscript. All authors read and approved the final manuscript.
Corresponding author
Ethics declarations
Competing interests
The authors declare that they have no competing interests.
Publisher’s Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Additional files
Additional file 1:
Relationship between setting stroke volume (sSV) and airway pressure amplitude (AMP) measured at the Ypiece. Legend: Markers indicate the sSV and mean AMP (n = 5). Standard deviation is not indicated. The results of the statistical significance test are as follows: BF = 10 vs BF = 20: P < 0.001 with sSV = 80–110 and 160–180, P < 0.05 with sSV = 120, ns with sSV = 130–150; BF = 10 or 20 vs BF = 30 or 40: P < 0.001 with all sSV; and BF = 30 vs BF = 40: ns with sSV = 80, P < 0.001 with sSV = 90–180. Dotted curves are secondorder approximation curves. Coefficient of correlations (R) and P values are as follows: BF = 10: R = 0.992, P < 0.001; BF = 20: R = 0.997, P < 0.001; BF = 30: R = 0.995, P < 0.001; and BF = 40: R = 0.990, P < 0.001. (DOCX 55 kb)
Additional file 2:
Actual stroke volume (aSV) with targeted aSVs of 80, 100, and 120 mL. Legend: The bar graph indicates the mean aSV (n = 5), and the vertical bar indicates the standard deviation. There are no significant differences between BF from 10 to 60 L/min with all targeted aSVs. (DOCX 75 kb)
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
About this article
Cite this article
Nagano, O., Yumoto, T., Nishimatsu, A. et al. Bias flow rate and ventilation efficiency during adult highfrequency oscillatory ventilation: a lung model study. ICMx 6, 11 (2018). https://doi.org/10.1186/s4063501801763
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/s4063501801763